Detector for imaging of explosions: present status and future prospects   with higher energy X-rays
  The detector for imaging of explosions (DIMEX) is in operation at the synchrotron radiation (SR) beam-line at VEPP-3 electron ring at Budker INP since 2002. DIMEX is based on one-coordinate gas ionization chamber filled with Xe-CO2(3:1) mixture at 7atm, and active Frisch-grid made of Gas Electron Multiplier (GEM). The detector has spatial resolution of ~0.2mm and dynamic range of ~100 that allows to realize the precision of signal measurement at a percent level. The frame rate can be tuned up to 8 MHz (125 ns per image) and up to 32 images can be stored in one shot. At present DIMEX is used with the X-ray beam from 2T wiggler that has ~20 keV average energy. Future possibility to install similar detector at the SR beam-line at VEPP-4 electron ring is discussed. 
V. M. Aulchenko, O. V. Evdokov, L. I. Shekhtman, K. A. Ten, B. P.   Tolochko, I. L. Zhogin, V. V. Zhulanov