An Automated Laboratory Laser Heating Arrangement for Materials   Synthesis at High Temperatures and High Pressures
  This paper describes the automation of a laser heating arrangement for synthesizing and studying materials at high pressures (up to ~ 1 Mbar) and high temperatures (up to ~ 5000 K). In this arrangement, a diamond anvil high-pressure cell (DAC) containing a microscopic sample of typical diameter ~50-100 micrometer, is mounted on a precision X-Y nanomotor stage that forms part of an IR laser heating optical assembly. Automation of this stage has been accomplished using a LabVIEW virtual instrument program to manipulate the X and Y stages using nanopositioning systems. This has a major feature of enabling a rastered heating of the sample over a user-defined circular area, without any operator intervention in addition to a virtual joystick to position the sample with respect to the laser spot. This auto-rastering feature has the advantage of offering uniform exposure of a circular area of the sample to the incident heating laser beam apart from drastic reduction in scan time compared to a manual scan. The diameter of the circle can be varied from a maximum of ~24 mm down to the focal spot size of the laser (~few micrometers), enabling thereby usage of the laser heating arrangement for heating microscopic samples under high-pressure in a DAC, as well as bulk samples at atmospheric pressure. Examples for both macro and micro scale automated laser-heating experiments have been presented. In particular, at the micro scale, auto-raster heated carbon samples at ~ 17 GPa and ~2000 K showed excellent signatures of diamond formation compared to manually raster heated samples, highlighting the unique advantage of auto-raster heating. 
N. Subramanian, Mrithula Ummru, R. Bindu, N. R. Sanjay Kumar, M.   Sekar, N. V. Chandra Shekar, and P. Ch. Sahu