Design of a scanning gate microscope in a cryogen-free dilution   refrigerator
  We report on our design of a scanning gate microscope housed in a cryogen-free dilution refrigerator with a base temperature of 15 mK. The recent increase in efficiency of pulse tube cryocoolers has made cryogen-free systems popular in recent years. However, this new style of cryostat presents challenges for performing scanning probe measurements, mainly as a result of the vibrations introduced by the cryocooler. We demonstrate scanning with root-mean-square vibrations of 0.8 nm at 3 K and 2.1 nm at 15 mK in a 1 kHz bandwidth with our design. 
Matthew Pelliccione, Adam Sciambi, John Bartel, Andrew Keller, David   Goldhaber-Gordon
physics.ins-det cond-mat.mes-hall