The HERMES Back Drift Chambers
  The tracking system of the HERMES spectrometer behind the bending magnet consists of two pairs of large planar 6-plane drift chambers. The design and performance of these chambers is described. This description comprises details on the mechanical and electronical design, information about the gas mixture used and its properties, results on alignment, calibration, resolution, and efficiencies, and a discussion of the experience gained through the first three years of operation. 
S. Bernreuther et al