Polarimetry with NICMOS
  NICMOS cameras 1 and 2 each carry a set of three polarizing elements to provide high sensitivity observations of linearly polarized light. The polarizers are bandpass limited and provide diffraction-limited imaging in camera 1 at 0.8 - 1.3um, and in camera 2 at 1.9-2.1um. The NICMOS design specified the intra-camera primary axis angles of the polarizers to be differentially offset by 120 degree, and with identical polarizing efficiency and transmittance. While this ideal concept was not strictly achieved, accurate polarimetry in both cameras, over their full (11" and ~19.2" square) fields of view was enabled through ground and on-orbit calibration of the as-built and HST-integrated systems. The Cycle 7 & 7N calibration program enabled and demonstrated excellent imaging polarimetric performance with uncertainties in measured polarization fractions <=1%. After the installation of the NICMOS Cooling System (NCS), the polarimetric calibration was re-established in Cycle 11, resulting in systemic performance comparable to (or better than) Cycle 7 & 7N. The NCS era NICMOS performance inspired the development of an earlier conceived, but non-implemented, observing mode combining high contrast coronagraphic imaging and polarimetry in camera 2. We successfully executed a program to calibrate and commission the "Coronagraphic Polarimetry'' mode in NICMOS in Cycle 13, and the mode was made available for GO use in Cycle 14. We discuss the data reduction and calibration of direct and coronagraphic NICMOS polarimetry. Importantly, NICMOS coronagraphic polarimetry provides unique access to polarized light near bright targets over a range of spatial scales intermediate between direct polarimetry and ground-based (coronagraphic) polarimetry using adaptive optics. 
D.C. Hines (1), G. Schneider (2) ((1) Space Science Institute, (2)   Steward Observatory, The University of Arizona)