Signal Reconstruction for the MAGIC Telescope
  Until April 2007 the MAGIC telescope used a 300 MSamples/s FADC system to sample the shaped PMT signals produced by the captured Cherenkov photons of air showers. Different algorithms to reconstruct the signal from the read-out samples (extractors) have been implemented and are described and compared. Criteria based on the obtained charge and time resolution/bias are defined and used to judge the different extractors, by applying them to calibration, cosmic and pedestal signals. The achievable charge and time resolution have been derived as functions of the incident number of photo-electrons. 
J. Albert, et al. (for the MAGIC Collaboration)