Status of the XMM-Newton cross-calibration with SASv6.5.0
  Further achievements of the XMM-Newton cross-calibration - XMM internal as well as with other X-ray missions - are presented. We explain the major changes in the new version SASv6.5 of the XMM-Newton science analysis system. The current status of the cross-calibration of the three EPIC cameras is shown. Using a large sample of blazars, the pn energy redistribution at low energy could be further calibrated, correcting the overestimation of fluxes in the lowest energy regime. In the central CCDs of the MOSs, patches were identified at the bore-sight positions, leading to an underestimation of the low energy fluxes. The further improvement in the understanding of the cameras resulted in a good agreement of the EPIC instruments down to lowest energies. The latest release of the SAS software package already includes corrections for both effects as shown in several examples of different types of sources. Finally the XMM internal cross-calibration is completed by the presentation of the current cross-calibration status between EPIC and RGS instruments. Major efforts have been made in cross-calibrations with other X-ray missions, most importantly with Chandra, of course, but also with currently observing satellites like Swift. 
M. Stuhlinger, B. Altieri, D. Baskill, K. Dennerl, M.P. Esquej, M.   Freyberg, F. Haberl, J.W. den Herder, J. Kaastra, M.G.F...