An effective thiol-reactive probe for differential scanning fluorimetry with a standard RT-PCR device
Differential scanning fluorimetry (DSF) is used to assess protein stability, transition states, or the Kd\xe2\x80\x99s of various ligands, drug molecules and antibodies. All fluorescent probes published to date are either incompatible with hydrophobic proteins/ligands, which precludes analyses of transmembrane or membrane associated proteins, or have excitation and detection wavelengths outside the range of RT-PCR machines, necessitating the use of dedicated devices. Herein, we describe a thiol-reactive probe BODIPY FL L-cystine (BFC) to overcome both of these shortcomings. The probe supports an inexpensive application of DSF measurements suitable for detection with standard RT-PCR machines in a hydrophilic or hydrophobic environment.
Introduction<!>Preparation and storage of the BFC probe<!>Preparation and storage of the Sypro-orange probe<!>Preparation of the lysozyme samples<!>Purification of bovine rhodopsin (UNIPROT ID: P02699)<!>Experimental sample preparation and fluorescence measurements<!>Data analysis<!>BODIPY FL L-cystine method validation<!>Rhodopsin stability<!>Discussion<!>Conclusions
<p>Differential scanning fluorimetry (DSF) has versatile applications in protein biochemistry, (membrane-) protein crystallography, electron microscopy, as well as ligand, drug and antibody screening (1–12). Thermal denaturation of a protein can be monitored with different fluorescent probes that interact with exposed hydrophobic moieties or Cys residues. The change of fluorescence upon interaction is monitored by a UV/ VIS- detector and is plotted as a function of temperature (4). The response curves then can be used to determine the melting temperature of the protein calculated as the first derivative of the sigmoidal curve, or by fitting the curve to the Boltzmann equation (4, 5, 13, 14).</p><p>Herein, we introduce the boron-dipyrromethene (BODIPY) based probe BFC which enabled DSF measurements of various proteins in a hydrophobic environment and detection of the fluorescent signal (λex = 504 and λem = 511 nm) with a standard RT-PCR machine. The functionality of the BFC probe as a reporter dye was assessed with chicken lysozyme C (UNIPROT ID: P00698) and bovine rhodopsin (UNIPROT ID: P02699), as a representative membrane protein.</p><!><p>One mg of BFC was purchased from Invitrogen (B-20340). The probe was diluted to a final concentration of 10 mM in DMSO and stored in 1 μl aliquots in the dark at −20 °C. The final working concentration of 2 μM BFC was used throughout the DSF experiments by diluting the DMSO stock with deionized water before adding it to a sample.</p><!><p>The Sypro-orange 5000x concentrate was purchased from Invitrogen (S-6650), and stored in the dark at 4 °C. A final working concentration of 5x Sypro-orange was used throughout the DSF experiments by adding the DMSO stock solution to the sample (6). If the required volume was less than 1 μl, 5000x Sypro-orange was diluted in deionized water before adding it to a sample.</p><!><p>Lysozyme C (UNIPROT ID: P00698) was purchased from Sigma Aldrich (L6876). Lyophilized lysozyme powder was solubilized in PBS buffer (NaCl, 137 mM; KCl 2.7 mM; Na2HPO4, 10 mM; and KH2PO4, 1.8 mM) pH 7.4 to a final concentration of 10 mg/ml. A series of two-fold dilutions in PBS was prepared in a MicroAmp Fast Optical 96-Well Reaction Plate (4346906) from Applied Biosystems.</p><!><p>All manipulations were carried out in a darkroom under dim red illumination (>650 nm). Rod outer segments (ROS) were prepared as previously described (15, 16). Briefly, crude ROS were isolated from dark adapted bovine retina (W. L. Lawson Co., Lincoln, NE) by a sucrose gradient extraction method. Purified ROS membranes (20 mg/mL rhodopsin) were solubilized by the zinc/alkyl-glucoside extraction method and centrifuged at 100,000g for 40 min to extract rhodopsin (17). The clear supernatant was further purified by gel filtration chromatography on a Superdex-200 column (GE-Healthcare) in 50 mM Tris, pH 7.5, 100 mM NaCl, 1 mM MgCl2 and 0.02% n-dodecyl β-D-maltoside. Purified rhodopsin was concentrated to 2.6 mg/ml and used for DSF experiments.</p><!><p>The dilution series were constructed in duplicate. Each well contained 9 μl of protein sample plus 1 μl of dye. The plate was sealed with ClearSeal film (HR4-521) from Hampton Research. The final probe concentration was 2 μM for BFC and 5x times for Sypro-orange. The plate was incubated for 10 min on ice prior to DSF measurements. The dilution series revealed the optimal working concentration for DSF measurements, which was then verified in quadruplicate.</p><p>All DSF measurements were performed with a StepOnePlus System (4376600) from Applied Biosystems. The melting curve experiments were conducted and recorded using StepOne Software v2.3 from Applied Biosystems. The SYBR®, FAM™ and ROX™ channels were used simultaneously to record the change in fluorescence. The run was set to cool down to 4 °C within 10 s, kept at 4 °C for 1 min and then increased 1 °C per min in a step and hold manner up to 99.9 °C. The multicomponent data was exported on a Microsoft EXCEL sheet and analyzed with the XLfit package 5 (5).</p><!><p>Data were analyzed by averaging and normalizing the fluorescence intensities from the multicomponent experiments. Normalized curves were fitted to the Boltzmann equation (i) using the XLfit package 5.</p><!><p>The melting temperature of lysozyme (74 ± 0.5 °C) w as previously determined using different methodology (18–21). A comparison between Sypro-orange and BFC was made with a lysozyme concentration at 2.25 mg/ml (Fig. 1a & b). The melting temperatures were 68.6 °C with Sypro-orange (Fig. 1b) and 74.4 °C with the BF C probe (Fig. 1d). The melting temperature obtained with BFC coincides with label-free measurements as stated above. Conversely, measurements with the Sypro-orange probe differed by 5.8 °C. This difference originates from the pH dependence of the lysozyme DSF data measured with the Sypro-orange probe as reported previously (21). Agreement of the melting temperature between label free measurements and BFC supports the usage of this probe for DSF.</p><!><p>The melting temperatures of bovine rhodopsin in ROS disk membranes are 71.9 °C and 55.9 °C for the dark and bleached states, respectiv ely (22, 23). The measurements with Sypro-orange could not be analyzed because of its high initial fluorescence and the absence of a transition (Fig. 2a & b). Figure 2b reveals the averaged quadruplicate DSF measurements at 9.14 μg/ml sample concentration. Analysis of this sample revealed a melting temperature of 72.1 °C, which is consistent with the published value (Fig. 2c). Overall, the data show that BFC is an effective probe to obtain DSF measurements of hydrophobic proteins.</p><!><p>Comparing DSF measurements between Sypro-orange and BFC demonstrated the superior data obtained with BFC. Moreover, the melting temperature of lysozyme measured with BFC is identical with the results obtained with label-free methods. The concentration-dependent melting temperatures observed with both lysozyme and rhodopsin indicated that DSF measurements have to be conducted at a fixed protein concentration to allow accurate comparison.</p><p>BFC belongs to the class of thiol-reactive probes, and therefore it suffers from similar shortcomings as CPM (9). The main limitations are the pH dependence of thiol formation, the interaction with reducing agents and the interference of the probe with a ligand binding site (4, 9, 14, 24). Nevertheless, we have shown these shortcomings can be overcome by careful sample preparation, initial evaluation of the protein, probe concentrations, and careful data analyses.</p><!><p>We demonstrate here that BFC is a versatile thiol-reactive compound that can be used for high-throughput DSF measurements of membrane proteins, proteins with hydrophobic binding sites or hydrophobic ligands. The absorption and fluorescent emission of BFC allows signal detection with a standard RT-PCR device. Moreover, the initial dimeric state of BFC enhances the intensity of the fluorescent signal, producing greater assay sensitivity and permitting lower probe concentrations, if desired. These properties render BFC a superior probe for high-throughput DSF measurements with standard RT-PCR equipment.</p>