Discussion Chain ID: 85679
================================================================================

User asked: No response from instrument via serial communication

I am using a Teensy 3.6 to communicate with an external instrument via a serial connection. However, I am not able to get the instrument to respond to my commands. The idea is to send a command from the computer via the serial monitor to the Teensy, which then can send it to the instrument (I know that the Teensy is not needed with this configuration but this is just for testing of a more complex set-up). The instrument uses an RS232 connector so I use this [RS232 shifter](https://www.sparkfun.com/products/449) to connect it to the Teensy. The serial communication settings of the instrument are: 38.4k baud rate, 8 data bits, 1 stop bit, no parity and a carriage return character should mark the end of the command string. A simple example of my code to read the input from the serial monitor and send a command to the instrument: #define FILTser Serial1 char Command; void setup() { Serial.begin(9600); FILTser.begin(38400); Serial.println("<Teensy is ready>"); } void loop() { Command = 'X'; recvCmd(); // Start sampling command if (Command == 'S') { FILTser.write("pumpctl=1\r"); FILTser.write('\n'); Serial.println("Sampling started"); } } void recvCmd() { if (Serial.available() > 0) { Command = Serial.read(); } } If I read the output from the Teensy on the Serial1 port with Putty I get the message correctly. If I send the same command directly to the instrument with Putty or Python I get the instrument to respond as I expect. This means that the instrument is working and the serial output from the Teensy is also fine so I really don't understand what could be wrong. I know that the instrument is not responding because I would expect a pump to turn on when it receives the command. Additionally, I tried to read the answer of the instrument on the serial port and I was receiving a `-1` but I would expect `OK` for a good command and `ERR` if the command is not recognized. To read the instrument answer on the serial port I tried using both a single character with `Response = FILTser.read();` and also a full string using the code from [here](https://forum.arduino.cc/t/serial-input-basics-updated/382007). An example of the Python code that I use to send commands to the instrument is the following: import datetime import serial import io import time # ============================================================================= # Configure serial port # ============================================================================= ser = serial.Serial() ser.baudrate = 38400 ser.port = 'COM5' ser.timeout = 1 #%% Define FILT functions def FILT_send(commnd): ser.write(commnd.encode('ascii')) time.sleep(0.5) value_read = ser.readline().decode('ascii') return(value_read) def FILT_control(): global q user_input = input('Type:\n \ - S to start sampling\n \ - E to end sampling\n \ - Q to quit the program\n') if user_input == 'S': FILT_send('pumpctl=1\r') print('Sampling started') if user_input == 'E': FILT_send('pumpctl=0\r') print('Sampling ended') if user_input == 'Q': print('The end') q = False #%% Main ser.open() q = True while q: FILT_control() ser.close()

Discussion:
Comment: Comments are not for extended discussion; this conversation has been [moved to chat](https://chat.stackexchange.com/rooms/128792/discussion-on-question-by-baccandr-no-response-from-instrument-via-serial-commun).

Answer 1 (3 votes):
The Sparkfun RS232 shifter is not a great device. It relies on whatever it's connected to (on the 9-pin port) providing the negative rail for the EIA-232 voltage levels. That's fine when you connect to something that _has_ those voltages available, such as a computer, but many "devices" don't - they will themselves have a similar shifter inside and will rely on whatever _they_ are plugged in to for providing the needed negative voltage. Put the two together and _no-one_ will provide that negative voltage. Instead you should use a MAX232 (or MAX3232 for 3.3V operation) based converter. This has a built-in voltage doubler and voltage inverter to convert the 5V into +/-10V to give a proper EIA-232 compliant signal which the device should then be happy with.


================================================================================

