Discussion Chain ID: 29533
================================================================================

User asked: Sum values, store them and print the last value

I have a mass flow meter connected to an Arduino Uno board. I have managed to read the volume flow in L/hour, but now I'm struggling to implement a code that stores the values so that I see the number of Liters used. For instance, I can see on my monitor this: 0 L/hour 0 L/hour And if I have some water running through, 10 L/hour 20 L/hour 20 L/hour Then when I stop the flow it goes back to 0. 0 L/hour Is there any ways to sum up time instances, get the total number of liters and keep it printed during every instance? Mathematically is not that difficult (Volume = Volume Flow in L/hour * time), but I don't know how to implement this. Can anyone help with this? It can be in seconds or ms or whatever. Paul

Answer 1 (4 votes):
What you want to do is _integrate_ the flow rate vs time curve. You have to keep track of the current flow rate. You also need to choose some unit of time measurement; this depends on how often you're sampling the flow rate. For instance, if you sample it every second, then you should convert your L/hr flow rate to L/s. This means a flow rate of 10L/hr is equivalent to 10L/3600s = 0.002778 L/s. So, if you get a reading of 10 L/hr for 10 seconds, `0.002778 L/s * 10 s = 0.02778 L` of water flowed through your valve or whatever. An accumulating sum is what you need. You can try something like this: #define INTERVAL 1000 // sampling interval in millisec #define HR_TO_SEC 2.778E-4 // number of hours in a second int curr_rate = 0; // current flow rate in L/hr unsigned long lastRead = 0; // the last time the rate was sampled float totalVolume = 0; void loop(){ if (millis() - lastRead >= INTERVAL){ lastRead += INTERVAL; curr_rate = getFlowRate(); // or whatever function you call; should take << 1 sec totalVolume += curr_rate * HR_TO_SEC; // update the volume Serial.println(totalVolume, 4); } } This rectangular method is reasonably accurate, depending on your application. If you want a better approximation, google "trapezoidal numerical integration" or "Simpson's method" if you want even better approximation.

Discussion on this answer:
Comment: The higher order integration methods achieve better accuracy by leveraging the _smoothness_ of the integrand. Here, since there is a valve, the flow is unlikely to be smooth. If you want better accuracy, you only have to sample at shorter intervals.
Comment: Also, I replaced `lastRead = millis();` by `lastRead += INTERVAL;`. Otherwise you have a systematic bias, as the average sampling interval would be larger than `INTERVAL`.
Comment: Thank you both for the help. I have managed to implement this code and it does indeed store the data. I have poured 1 L of water through the meter and the monitor reads 1.2 L. Not a huge difference, but I will change the method of integration and we will see if I can correct that. Also, I had to remove "unsigned" for "float totalVolume = 0;". The code would give an error otherwise.
Comment: @Paul You can do as Edgar Bonet says and reduce the interval to 200 ms maybe. Also take care to change HR_TO_SEC to reflect that: convert your interval to hours. If this solves your problem, consider accepting it.
Comment: Okay. I will do some experiments these days and see how it goes. I will write back! Thank you

Answer 2 (1 votes):
This will be helpful. reading is your flow meter value in L/Hour float reading;//value of the flow meter long startTime; float total=0; void setup() { Serial.begin(9600); startTime=millis(); } void loop() { //read your flow meter to reading variable. total+=reading*float((millis()-startTime))/3600000.0; Serial.println(total); }

Discussion on this answer:
Comment: Thank you very much! I have tried with this code and it keeps giving me 0. Also, I needed to add long startTime=millis(); instead of simply declaring long startTime and then calling it. Is the "reading" function responsible for me getting 0? Does it have to be related to the signal or the pinmode?
Comment: reading is the variable..You must pass your flow meter reading to that variable as a float type. you must enter it before Serial.println() function as I commented. if you can give previous code,I will be able to give full code.
Comment: The code can be found here: http://forum.arduino.cc/index.php?topic=8548.0 (It is too long to post it here). Thank you!

Answer 3 (1 votes):
In a recent comment, you posted a link to a [forum thread](http://forum.arduino.cc/index.php?topic=8548.0) with the code for reading the sensor. You should have provided this information up front, because it completely changes the problem. It appears that the flow meter sends pulses at a frequency proportional to the volume flow, and the code uses an interrupt to count the pulses over a specified time. TisteAndii's answer suggests you _integrate_ the flow readings to get a volume. And that suggestion makes perfect sense given the limited information he had available at the time. Now that we know how the flow meter is read, it makes no sense anymore: * the interrupt service routine counts a number of pulses that is proportional to the _total volume_ that has gone through the meter * the linked code _differentiates_ that reading in order to get the flow * you are now _integrating_ that flow in order to get the total volume. Instead of doing an integration on top of a differentiation, you could just convert the raw reading into a volume. Then you do not have to worry about any approximations done while differentiating or integrating. In the forum thread it is stated: > Pulse frequency (Hz) in Horizontal Test= 7.5Q, Q is flow rate in L/min. Writing this in a more mathematically accurate form (i.e. with unit-correctness) gives: > _f_ /Hz = 7.5 _Q_ /(L/min) where _f_ is the frequency and _Q_ the volume flow. Given that 7.5 Hz⋅min = 450 pulses, the above equation can be rewritten as > _f_ = 450 pulses/L × _Q_ or, in terms of integrated quantities: > volume = (pulse count) / (450 pulses/L) Here is an example code that just convert the pulse count to a volume: // Calibration constant of the flow meter. const float calibration = 1/450.0; // 450 pulses per liter // How often to print the measured volume. const uint32_t print_interval = 1000; // once per second // Number of pulses counted so far. volatile uint32_t pulse_count; // Count the pulses inside an ISR. void count_pulse() { pulse_count++; } // Return volume reading, optinally resetting the count to zero. float totalVolume(bool reset = false) { noInterrupts(); uint32_t pulse_count_copy = pulse_count; if (reset) pulse_count = 0; interrupts(); return pulse_count_copy * calibration; } void setup() { pinMode(2, INPUT_PULLUP); attachInterrupt(0, count_pulse, RISING); Serial.begin(9600); } void loop() { static uint32_t last_print; if (millis() - last_print >= print_interval) { last_print += print_interval; Serial.println(totalVolume()); } } Note that you may want to reset the count from time to time. I added an optional parameter to `totalVolume()` for this purpose. If you never reset the counter, it will overflow after approximately 9544 m3.


================================================================================

