Discussion Chain ID: 70857
================================================================================

User asked: Logging PAR with quantum sensor and Arduino

I want to link a PAR quantum sensor (either Licor or Skye Instruments) to Arduino to log PAR, in order to measure light in a forest. I have an Arduino Uno + data logging shield (with RTC, SD card and ADS1115) and have connected the PAR sensor via A0 to measure differential voltage (A0 –– S+ | S- –– GND | A1 –– GND). Code as follows: ads.setGain(GAIN_TWOTHIRDS); adc0 = ads.readADC_Differential_0_1(); Voltage = (adc0 * 6.144)/pow(2, 15); Voltage = Voltage*1000; // conversion to millivolts When I wire the Arduino to itself to measure a differential voltage (A0 –– 3V and A1 –– GND), it measures beautifully. Furthermore, when I use a voltmeter to test the Voltage across the + - wires from the sensor, I get sensible readings. Put the two together – nonsense. I get noisy V readings that don't make sense. When I cover the sensor my V is actually going up, not down as it should! I have tried changing the GAIN, but unfortunately this makes no difference to the problem. I have been looking at [PARduino](https://academic.oup.com/treephys/article/34/6/640/2338115) for help/inspiration, but they include an amplifier in their setup, which is expensive and I'm not sure is necessary / would solve my problem. Very new to Arduino. Any and all advice would be very much appreciated.

Discussion:
Comment: Could you connect the sensor to the Arduino and, at the same time, to the voltmeter? Do you still get sensible readings from the voltmeter if you do so?
Comment: Please provide a link to the datasheet of the used sensor. I'm not sure, if you can simply omit the amplifier
Comment: Thanks for your comments Edgar and chris. The problem must have been arising from a faulty board or some such. In the end, I was able to wire up the sensor as described above, using a more fleshed-out version of the code. No amplifier necessary, you can just adjust the GAIN. I then ran my Arduino sensor alongside a Skye Instruments SpectroSense 2+ to 'calibrate' it. Plotting the Arduino voltages against the SpectroSense PAR values gave quite a nice fit! Apologies for the long delay in replying – was on fieldwork.


================================================================================

