Discussion Chain ID: 96599
================================================================================

User asked: Interpreting Analog Output from Aalborg Mass Flow Meter GFM

I am attempting to connect [this](https://www.aalborg.com/gfm-mass-flow-meter_o/20) device to my computer via RS232. I am debating the simplest way of going about this and was inquiring if my Uno R3 could come into use, turning the analog symbols of 0-5 Vdc and 4-20 mA from the GFM into an output in serial port over a COM port. I have an RS232 to USB and I am just now tasked with completing this. [](https://i.sstatic.net/MBzO68xp.png) [](https://i.sstatic.net/oJnusfIA.png)

Discussion:
Comment: `0-5 Vdc and 4-20 mA` ... it's `0-5 Vdc or 4-20 mA` ... they both carry the same information
Comment: what is your question? ... please add a focused, answerable question to your post
Comment: @jsotola - Thanks for highlighting the cross-site multi-posting of this question on EE and here. The [copy on EE](https://electronics.stackexchange.com/q/720547) has been [closed](https://meta.stackexchange.com/q/64068), so this one can continue without duplication of effort across the sites.
Comment: Welcome to SE/Arduino! Please take the [tour] to learn how this site works. Then come back and [edit] your question to clarify. -- Why did you tag with `modbus`? -- Are you aware that your Arduino is able to measure voltages? What stops you from using `analogRead()`? -- If you have an attempt of a sketch, please add it. And why it does not work for you. Make sure to format it as source. See the help of the edit box for details.

Answer 1 (0 votes):
Although it is common for mass flow meters to have a serial interface (which could well be Modbus), this particular model has only analog outputs. If you have an Uno at hand, your simplest option is to use it for reading the 0–5V meter output, and send the data to the PC through the Arduino's serial-over-USB connection. Connect: * the meter's pin 3 (0 to 5 VDC common) to the Arduino's GND * the meter's pin 2 (0 to 5 VDC output indication) to the Arduino's A0 * the Arduino's USB connector to your PC You don't need the RS-232 to USB converter, as you will not be using RS-232 at all. The Arduino will simply read the analog voltage, optionally convert it to flow units, and sent that through it's serial port: const float flow_range = ...; // the meter's full scale const uint8_t input_pin = A0; // analog input used const uint32_t print_period_ms = 200; // tune to taste void setup() { Serial.begin(9600); } void loop() { float flow = analogRead(input_pin) * (flow_range / 1024); Serial.println(flow); delay(print_period_ms); }


================================================================================

