Discussion Chain ID: 29511
================================================================================

User asked: Results for GP2Y1010AU0F Dust Sensor

I am using Dust Sensor GP2Y1010AU0F I downloaded the code from the manufacturer here: <http://www.waveshare.com/wiki/Dust_Sensor> /********************************************************************************************************* * * File : DustSensor * Hardware Environment: * Build Environment : Arduino * Version : V1.0.5-r2 * By : WaveShare * * (c) Copyright 2005-2011, WaveShare * http://www.waveshare.net * http://www.waveshare.com * All Rights Reserved * *********************************************************************************************************/ #define COV_RATIO 0.2 //ug/mmm / mv #define NO_DUST_VOLTAGE 400 //mv #define SYS_VOLTAGE 5000 /* I/O define */ const int iled = 7; //drive the led of sensor const int vout = 0; //analog input /* variable */ float density, voltage; int adcvalue; /* private function */ int Filter(int m) { static int flag_first = 0, _buff[10], sum; const int _buff_max = 10; int i; if(flag_first == 0) { flag_first = 1; for(i = 0, sum = 0; i < _buff_max; i++) { _buff[i] = m; sum += _buff[i]; } return m; } else { sum -= _buff[0]; for(i = 0; i < (_buff_max - 1); i++) { _buff[i] = _buff[i + 1]; } _buff[9] = m; sum += _buff[9]; i = sum / 10.0; return i; } } void setup(void) { pinMode(iled, OUTPUT); digitalWrite(iled, LOW); //iled default closed Serial.begin(9600); //send and receive at 9600 baud Serial.print("*********************************** WaveShare ***********************************\n"); } void loop(void) { /* get adcvalue */ digitalWrite(iled, HIGH); delayMicroseconds(280); adcvalue = analogRead(vout); digitalWrite(iled, LOW); adcvalue = Filter(adcvalue); /* covert voltage (mv) */ voltage = (SYS_VOLTAGE / 1024.0) * adcvalue * 11; /* voltage to density */ if(voltage >= NO_DUST_VOLTAGE) { voltage -= NO_DUST_VOLTAGE; density = voltage * COV_RATIO; } else density = 0; /* display the result */ Serial.print("The current dust concentration is: "); Serial.print(density); Serial.print(" ug/m3\n"); delay(1000); } My results seems rounded or always returning in certain 'steps'. (below are my results) Is there a rounded error in the code or something else that could be going on? The current dust concentration is: 134.84 ug/m3 The current dust concentration is: 134.84 ug/m3 The current dust concentration is: 134.84 ug/m3 The current dust concentration is: 134.84 ug/m3 The current dust concentration is: 134.84 ug/m3 The current dust concentration is: 134.84 ug/m3 The current dust concentration is: 134.84 ug/m3 The current dust concentration is: 124.10 ug/m3 The current dust concentration is: 124.10 ug/m3 The current dust concentration is: 124.10 ug/m3 The current dust concentration is: 113.36 ug/m3 The current dust concentration is: 113.36 ug/m3 The current dust concentration is: 124.10 ug/m3 The current dust concentration is: 124.10 ug/m3 The current dust concentration is: 113.36 ug/m3 The current dust concentration is: 113.36 ug/m3 The current dust concentration is: 113.36 ug/m3 The current dust concentration is: 124.10 ug/m3 The current dust concentration is: 124.10 ug/m3 The current dust concentration is: 124.10 ug/m3 The current dust concentration is: 134.84 ug/m3 The current dust concentration is: 134.84 ug/m3 The current dust concentration is: 124.10 ug/m3

Answer 1 (1 votes):
Its because the resolution of the analog read is 10 bits or 1024 discrete values. In your code the density is basically calculated as: density = (((5000 / 1024.0) * analog_value * 11) - 400) * 0.2 You can calculate the resolution as: (5000 / 1024.0) * 11 * 0.2 and you get `10.7421875`, which matches your observation. If your filtering (calculating the average over last 10 values), would calculate and return a float, you would get smoother values.


================================================================================

