Discussion Chain ID: 44905
================================================================================

User asked: Reading pressure sensor using Arduino analog

I'm using a Arduino Uno to read pressure and I want to know if I have wired/coded the sensor correctly. I'm not sure I'm getting the correct readings (it could be the right readings - I just want to check I have done everything right). The sensor I am using is an OsiSense™ XMLP pressure sensor that reads between 0 and 100 bars and gives out a 0v (zero bars) to 10v (100 bars) charge. Due to the Arduino Uno only being capable of reading up to 5 volts, I've used a voltage divider to reduce the maximum voltage the Arduino receives to 5v. I've attached a hand-written diagram. Let me know if you need any more information. I've attached the wiring and also the code. [](https://i.sstatic.net/GRDyi.jpg) /* ReadAnalogVoltage from a OsiSense XMLP presure sensor Reads an analog input on pin 0, converts it to voltage and then bar pressure, and prints the result to the Serial Monitor and also MQTT. Graphical representation is available using Serial Plotter (Tools > Serial Plotter menu). Todo Static ip. set it to request IP from network Work out best way to average readings */ #include <SPI.h> //Serial Peripheral Interface (SPI) #include <Ethernet.h> // These libraries are designed to work with the Arduino Ethernet Shield (Ethernet.h) #include <PubSubClient.h> // This library provides a client for doing simple publish/subscribe messaging with a server that supports MQTT // sets the MAC address and IP address for the network byte mac[] = { 0xDE, 0xED, 0xBA, 0xFE, 0xFE, 0xED }; IPAddress ip(192,34,98,221); // IP Address of the client IPAddress server(192,34,98,201); // The IP address of the MQTT server void callback(char* topic, byte* payload, unsigned int length) { // handle message arrived } EthernetClient ethClient; PubSubClient client(server, 1883, callback, ethClient); void setup() { Serial.println("Begin setup"); Ethernet.begin(mac, ip); // Note - the default maximum packet size is 128 bytes. If the // combined length of clientId, username and password exceed this, // you will need to increase the value of MQTT_MAX_PACKET_SIZE in // PubSubClient.h // connects to MQTT and says hello world. Probably change this later on to something more appropriate if (client.connect("ArdRigTest")) { client.publish("outTopic","Ard Rig Test"); client.subscribe("inTopic"); } // start serial port Serial.begin(9600); } void loop() { // Note - the default maximum packet size is 128 bytes. If the // combined length of clientId, username and password exceed this, // you will need to increase the value of MQTT_MAX_PACKET_SIZE in // PubSubClient.h // read the input on analog pin 0: int sensorValue1 = analogRead(A0); delay(2); // read twice with small delay to get accurate reading. int sensorValue2 = analogRead(A0); Serial.println(sensorValue2); //Test code. using map function int MapBarPressure; MapBarPressure = map(sensorValue2, 0, 1023, 0, 100); if (client.connect("ArdRigTest" )) { client.publish("MapBarPressure",String(MapBarPressure).c_str()); //logs the pressure of the RigTest client.subscribe("MapBarPressure"); } delay(100); // Pauses the program for the amount of time (in miliseconds) specified as parameter. (There are 1000 milliseconds in a second.) client.loop(); } Thanks in advance.

Discussion:
Comment: Assuming that you are using analogRead() with 0 for 0V and 1023 for 5V it is a question of mapping the range 0-1023 to the pressure range 0-100 bar (map(0, 1023, 0, 100)). See https://www.arduino.cc/en/Reference/Map. Or use float to keep the resolution.
Comment: Which pressure sensor is it ? The output voltage is probably not 0.0 to 5.0V, but rather 0.5 to 4.5V. How good and stable is your Arduino 5V which is used to power the sensor ? That is important to know. Is the pressure sensor ratiometric ?
Comment: Its a OsiSense™ XMLP. It kicks out a 10v output so I've built a circuit with a voltage divider so the output to the Arduino is a maximum of 5V. The power source to the sensor is 24v and is stable.
Comment: I'm using analogread. I don't understand the maths behind map but I guess thats the beauty of the function. It simplifies the maths so I dont need to.
Comment: Using the map function is okay. Just use it. However, you have a problem with the accuracy. What is your voltage divider made of ? Please tell everything, the resistor values and so on. If that sensor was the 0.5-4.5 ratiometric version, then you would have a good accuracy. With the 0-10V sensor you don't. How good and stable is your Arduino 5V ? Do you need the full 100 bar range ? If you only need to measure up to 50 bar, then only a protection resistor would be needed.
Comment: 2 x 10k resistors. one leads to ground and the other leads to the Arduino analogue in. i don't think i need the the full 100 bar, i will know more after measuring the pressure. When you say how stable is the arduino 5v, can you expand on what you mean?
Comment: Please update your question. Rewrite it completely, mention the sensor, the resistors, and which Arduino board, and the result that you want (if you want a float value, then you can't use the map function). The sensor you have is not ratiometric and outputs a absolute voltage. When the Arduino reads that, the 5V of the Arduino is default used as a reference. If that 5V changes, so will the resulting measured pressure change, even if the sensor outputs the same voltage. Using the internal reference is better, but it can be 1.0 to 1.2 V and is different for every Arduino board.
Comment: Added all the details of the project as requested. Hope all the information is there for you. If not, ask away :-)
Comment: The Arduino and the sensor need to share a common ground: connect the negative terminals of the power sources together.

Answer 1 (1 votes):
If you need to use fractional pressure values, then the formula is float MapBarPressure; MapBarPressure = sensorValue2 / 10.24; This was obtained by seeing that the maximum reading from `AnalogRead(A0)` is `1024`, which corresponds to 5V on A0 (and which also means 10V output from the sensor.) If you read `1024` from A0, it means you are measuring 100bar, so bar = (A0 / 1024) * 100 Alternatively, you could maintain 1 decimal place of accuracy in bar if you keep with `int` values: int MapDeciBarPressure; MapDeciBarPressure = map(sensorValue2, 0, 1023, 0, 1000); Then your variable is holding "tenths of bar" (deciBar) and you only convert it to bar by dividing by 10.0 (not 10, which is an int, but 10.0, which is a float and does floating point math and returns a float response.) One other note: you write: // read the input on analog pin 0: int sensorValue1 = analogRead(A0); delay(2); // read twice with small delay to get accurate reading. int sensorValue2 = analogRead(A0); Just reading twice doesn't gain accuracy -- you have to average the values: int sensorValAvg = (sensorValue1 + sensorValue2) / 2; Then, use sensorValAvg for your less noisy value. **Edit:** Also, make sure you connect all ground lines together, both from the sensor and the Arduino!

Discussion on this answer:
Comment: The conversion factor is 1024, not 1023.
Comment: Caught me again! Edited.

Answer 2 (1 votes):
You can use `map`, ref. : <https://www.arduino.cc/en/Reference/Map> > Re-maps a number from one range to another. That is, a value of fromLow would get mapped to toLow, a value of fromHigh to toHigh, values in-between to values in-between, etc.

Discussion on this answer:
Comment: With the caveat that "The map() function uses integer math so will not generate fractions, when the math might indicate that it should do so. Fractional remainders are truncated, and are not rounded or averaged."
Comment: so sometimes it's worth thinking in decibars internally and only convert to bars when you need to show it to the user.
Comment: @josecanuc did the OP ask to get decimals?
Comment: Indeed the OP did not specify any requirements in that regard, but if he or she uses `map()`, I only wished to clarify the limitation.
Comment: Please use a more constructive way to explain it then!
Comment: Thanks for letting me know about integer, truncates and thanks for the map function :-) The truncations may cause us a problem, I may need more accuracy.
Comment: @resolver101 If you need accuracy I would use an op-amp scaled to 1/2 instead of the voltage divider. And make sure you use 1% or better resistors.

Answer 3 (0 votes):
Everything is just fine. All you have to do is to provide the common ground (0 potential) to your Arduino and your Sensor. The problem here is that they might be on a different potential. So just connect the ground of your 5 volts and 24 volts together and everything will work perfectly.

Answer 4 (0 votes):
**About accuracy.** I assume you have read @edgar-bonet about connecting the GND's. I assume you understand the other answer to calculate the pressure. I would write it like this: // This is for the DEFAULT 5V analog reference. // v = the voltage at the pin in Volts. // p = the pressure in bar. // Note that 'bar' is the pressure relative to the // normal atmospheric pressure at that moment. float v = (float) analogRead(A0) / 1024.0 * 5.0; float p = v / 5.0 * 100.0; The [`analogRead`](https://www.arduino.cc/en/Reference/AnalogRead) uses default the 5V as reference. That means that if the project is powered by a USB cable, the 5V could vary between 4.5 and 5.0 V. That 10% difference will be seen in the resulting calculated pressure. In that case you can not measure more accurate than 10%. If the project is powered with a power supply to the power barrel jack, then the resulting accuracy depends on the accuracy of the voltage regulator on the Arduino board. The analog reference can be set with [`analogReference`](https://www.arduino.cc/en/Reference/AnalogReference). The internal voltage reference of the Arduino Uno is 1.1V. That is reasonably stable, but it depends a little on the voltage of VCC and the temperature. The main problem is that it is different for each Arduino Uno, and it can be between 1.0 and 1.2 V. Therefor you have to measure it, and use the measured value in the calculation in the sketch. The input range is between 0V and the analog reference. Therefor you have to adjust the voltage divider to get 0..10V down to 0..1.1V. You could try: R1 = 100k and R2 = 10k. const int pressureSignalPin = A0; void setup() { analogReference( INTERNAL); } void loop() { // v = the voltage at the pin in Volts. // p = the pressure in bar. // Note that 'bar' is the pressure relative to the // normal atmospheric pressure at that moment. // The "(100.0 + 10.0) / 10.0" is the factor by the voltage divider // with 100k and 10k. // Measure the AREF pin, and fill the value in for "1.10". float v = (float) analogRead(A0) / 1024.0 * 1.10; float p = v * ((100.0 + 10.0) / 10.0); }

Discussion on this answer:
Comment: Thanks. I've connect the grounds between the arduino and the sensor so they are all common. I've not tried the internal reference yet.. The power in is from a POE converter to the usb cable. I've attached a USB current/voltage reader and you can see the video hear : https://www.youtube.com/watch?v=Tb7qJXcJl9M&feature;=youtu.be


================================================================================

