Discussion Chain ID: 3747
================================================================================

User asked: Strange values from LSM303 Magnetometer

I am having problems with values i get from Pololu MinIMU-9 boards's magnetometer. I am using Arduino-micro and the arduino library from Pololu. The sensor on the board is 3 axis accelero- and magnetometer LSM303. The problem is - the highest value always seems to be in vertical axis, not horisontal, as i would expect. For example, here is the data i got when i put sensor flat on the table and made 360 clockwise turn in horizontal plane: (values are X Y Z) -6 102 -516 -11 103 -515 -11 103 -515 -21 102 -514 -31 109 -515 -40 114 -512 -57 116 -509 -57 116 -509 -77 111 -507 -93 109 -506 -104 106 -507 -104 106 -507 -114 99 -508 -126 88 -504 -131 74 -507 -142 52 -507 -142 52 -507 -151 31 -506 -150 4 -509 -147 -27 -513 -147 -27 -513 -133 -60 -513 -119 -80 -519 -112 -87 -517 -102 -97 -521 -102 -97 -521 -89 -101 -522 -77 -101 -524 -50 -104 -527 -50 -104 -527 -43 -105 -529 -14 -98 -532 2 -92 -535 23 -92 -535 23 -92 -535 36 -81 -534 66 -54 -534 76 -32 -531 76 -32 -531 76 -21 -532 78 -7 -531 85 10 -528 91 39 -524 91 39 -524 94 50 -524 91 63 -524 76 75 -525 76 75 -525 63 82 -526 47 93 -525 44 98 -522 44 98 -522 38 104 -521 26 110 -521 26 112 -518 21 116 -519 21 116 -519 18 116 -518 12 120 -517 11 120 -515 11 120 -515 8 120 -517 5 122 -515 -5 123 -514 Turning the sensor "upside down" and then rotating it in horisontal plane again gives similar results, only Z axis has positive 500+ value. Shouldn't the magnetometer always point in the direction of the magnetic field - in X or Y direction? I have tried this in different rooms and buildings and got the same results. Rotating the sensor on other axes yields the same results - vertical axis has the largest (and constant) value. For reference, here is the code i used to get the values. #include <Wire.h> #include <LSM303.h> LSM303 compass; void setup() { Serial.begin(115200); Wire.begin(); compass.init(); compass.enableDefault(); } void loop(){ compass.readMag(); Serial.print(compass.m.x); Serial.print(" "); Serial.print(compass.m.y); Serial.print(" "); Serial.print(compass.m.z); Serial.print("\n"); delay(100); }

Answer 1 (5 votes):
The earth's magnetic field is not parallel with the ground. In some locations it can be pointing more down than across. NOAA has a [magnetic field calculator](http://www.ngdc.noaa.gov/geomag-web/#igrfwmm) where you can enter you latitude and longitude. For example, I am located at approx 19deg S 147degE and the field is North Comp East Comp Vertical Comp Total magnitude 32,434.4 nT 4,280.6 nT -36,035.1 nT 48,670.7 nT So the downwards strength is more than the other two combined for my location. I'm guessing the calculator will also show a very strong down component for your location. Some other things to be aware of are that the LSM303DLHC magnetometer can have quite large offsets, up to the equivalent of 2 earth magnetic fields, and therefore requires calibration. Also hard and soft iron distortions can affect the readings. Edit: Please see my answer [here](https://electronics.stackexchange.com/questions/22144/magnetometer-%E2%88%9E-shaped-calibration/22271#22271) for methods to calibrate the sensor.

Discussion on this answer:
Comment: Thenk you, this really seems to be the case here. `Horisontal intesity: 15,210.5 nT Vertical component: 49,455.8 nT`


================================================================================

