Discussion Chain ID: 23528
================================================================================

User asked: Erratic readings on Analogread with low voltages

I am using Arduino Uno for measuring sensor voltages. I am using multiplexer to switch between different sensors. The readings measured in the range of 800-900 mv are accurate, but at low millivolts (at about 20 mv) they are erratic. I read repeatedly to get average of 20 values. I get values such as 1.3, 1,4, 2,0, 3.8 etc. Can any one help on why it is happening and how to avoid it. Additional information: I am using reference voltage of 1.25 volt for better resolution. I tried having delay between two readings, though there is some improvement, still the problem persists. I appreciate the help by experts. Thanks

Discussion:
Comment: May be this is because the voltages that you are trying to measure are way below the noise floor and the arduino is measuring the noise from the pin.
Comment: Noise problems always need multiple approaches / attacks: 1) Scope it to see what is any real noise needs to be dealt with. 2) Dive into the Atmel spec to see if the ADC can actually make such a measurement. 3) Use exponential smoothing and play with the alpha constant. 4) Try a different power supply or try something clean like a battery.
Comment: You need a stable power-supply. You need a stable measuring voltage. You could try adding the small capacitor to the output of the thing you try to measure. You could try using the ADC_sleep mode on the ATMega328. This will sleep the MCU during the measurement, keeping noise down.
Comment: Addition to the other comments, if that low voltages are common in your project, maybe you are using wheat-stone bridges with strain gauges or similar sensors, it might be the best option to use a good low noise instrumental amplifier between the sensor and the multiplexer. This setup will give you manageable voltage levels.


================================================================================

