Discussion Chain ID: 39963
================================================================================

User asked: Problems with ACS758 Hall Effect Current Sensors - Calibration and no Signal

I purchased 3 ACS758 Hall Effect Current Sensors [datasheet](http://www.allegromicro.com/~/media/Files/Datasheets/ACS758-Datasheet.ashx) on ebay. A 50,100 and 200A, all three are bi-directional. They look like [this](https://www.tindie.com/products/BBTech/allegro-acs758-current-sensor-module-50a100a150a200a/). With **no** current the 50A outputs 477 on the analogue input (2.33v), the 100A outputs 509 (2.48v) and the 200A outputs 512 (2.50v). At no current they should all be at exactly mid range i.e. 512 (2.50v). The datasheet says they're factory calibrated as such. I've tested them one after the other, both with 5v from the Audrino and with a separate 5v supply. As I've tested one after the other it's not likely to be stray magnetism or VCC. I've then put them in a circuit drawing about 9amps. Accounting for the offset of zero amps the 50A and 100A measured the increased voltage correctly. The 200A doesn't change. Is there any other tests I can do to see if I'm doing something dumb or are they all damaged?

Discussion:
Comment: Thanks for the reply Nasha. I will try measure VIOUT with my cheap multimeter, I don't have a precision one. I (think) I understand your point about noise. With zero load I'm getting quite a consistent reading, just a different one from each device. It should be 2.50v from each device and instead it's 2.33v and 2.48v (according to the arudino). Even if they're not the correct voltages they're different to each other. The chip is supposed to come factory calibrated to 2.50v for zero load. Do you think noise or ADC accuracy could be the cause of the calibration problem?
Comment: Thanks again! For the calibration measurement I did take hundreds of readings and they were very consistent (within 1/1024). I checked VIOUT with my cheap multimeter and I got the same 2.33v and 2.48v. VCC was 4.98v. The board has a 0.1µF capacitor right next to the chip, I bought the whole board but just looked and they are.

Answer 1 (0 votes):
I contacted Allegro they were really helpful which was pleasantly surprising. The 200A has almost certainly failed. The 2.48v is within tolerance and the 2.33v is probably out. I've been able to better understand the datasheet sections on 0A calibration. "VIOUT(Q) may drift over the lifetime of the device by as much as ±25 mV." and there is an electrical offset voltage noted. @25C it's +/-5mv. It's +/-20mv between 0c and 50c. So still only 45mv. Even allowing for noise it's only 55mv. About 2.445V at zero amps would be the limit.

Discussion on this answer:
Comment: Annoyingly the 200A was the one with the perfect calibration but ......"I've then put them in a circuit drawing about 9amps. Accounting for the offset of zero amps the 50A and 100A measured the increased voltage correctly. The 200A doesn't change." .... The 200A was perfect at zero amps and stayed at 2.50v regardless of current.... completely broken.


================================================================================

