Discussion Chain ID: 3864
================================================================================

User asked: Parse data from sensor without delimeters

Im using a variant of Arduino Due embedded on a ARM (Udoo). And I have connected a Medlab SPO2 Pearl 200 sensor to Serial1. Im just trying to parse input from Serial1 to Serial, in a human-readable format (Hexadecimal). The protocol is described in this image: And using this Arduino code I reveice two times the "no finger detected line" int lineLength = 16; int spo2read[16]; int readed = 0; if (Serial1.available()) { int inByte = Serial1.read(); if (readed < lineLength) { spo2read[readed] = inByte; } readed++; if (readed == lineLength) { Serial.print("SPO2: "); for (int i = 0; i <= lineLength; i++) { Serial.print(spo2read[i], HEX); Serial.print( delimeter); } Serial.println(""); readed = 0; } } Output with no finger is: SPO2: A,FB,2,F9,0,FA,0,FC,A,FB,2,F9,0,FA,0,FC,0, Buf if I put my finger, all is messed up: SPO2: 7C,74,6B,62,5A,51,4A,42,3B,34,30,2E,2F,30,31,33,0, SPO2: 36,38,3B,3E,3D,3C,3C,3B,3B,38,34,2F,2C,2B,2B,29,0, SPO2: 24,21,1E,1D,1E,1E,1F,20,22,24,26,28,2B,2D,30,32,0, SPO2: 35,37,39,3C,3E,40,42,44,45,4A,4F,53,55,57,59,5C,0, SPO2: 5E,60,61,F9,0,FA,0,FC,1A,FB,0,F8,62,62,62,63,0, Sensor doesn't sends a \n or \r so I don't know exactly how to parse the data

Discussion:
Comment: It says there is a marker byte. You have to look for the marker byte to know when the message has ended.
Comment: as per http://www.medlab-gmbh.de/english/downloads/pearl100_144.pdf page 13 marker bytes and example C code page 16 are values greater than or equal than F0

Answer 1 (0 votes):
actually I don't think it is all mixed up as per geometrikal use the markers to parse the data stream I believe your above sample data started mid-stream during the 0xF8 of the wave form. F8,... ...,7C,74,6B,62,5A,51,4A,42,3B,34,30,2E,2F,30,31,33,0,36,38,3B,3E,3D,3C,3C,3B,3B,38,34,2F,2C,2B,2B,29,0,24,21,1E,1D,1E,1E,1F,20,22,24,26,28,2B,2D,30,32,0,35,37,39,3C,3E,40,42,44,45,4A,4F,53,55,57,59,5C,0,5E,60,61 F9,0 FA,0 FC,1A FB,0 F8,62,62,62,63,0,... I believe reading this as a finite length does not portray the data for easy interpetation. Given that 0xF8 is a variable length protocol. Where the reparse of your data initially appears to be legitimate parse. That said I do read the SPO2, Pulse as zero. Where it may be likely the sensor has not finished its reading and auto gain cycles.

Answer 2 (0 votes):
I have used the source code available on the Medlab web, and do some "modifications" to be Arduino compatible. I dont wanna the waveform data, because my final program will do the waveforms acording the data received, so I don't "listen" to the waveform packet boolean startbyte = false; byte buf = 0; while (Serial1.available() > 3) { buf = Serial1.read(); if (buf > 0xfc) { buf = Serial1.read(); } if ((startbyte == true) && buf > 0xF7) { startbyte = false; } if (buf > 0xf7 && startbyte == false) { //startbyte = true; if (buf == 0xf8) { startbyte = true; } if (buf > 0xf8) { byte RX = buf; buf = Serial1.read(); if (buf > 0xfc) { buf = Serial1.read(); } if (RX == 0xf9) { //SPO = buf; Serial.print("SV:"); Serial.print(buf, DEC); Serial.println(); } else if (RX == 0xfa) { //Pulse = buf; Serial.print("SPV:"); Serial.print(buf, DEC); Serial.println(); } else if (RX == 0xfb) { //Status = buf; Serial.print("StV:"); Serial.print(buf, DEC); Serial.println(); } else if (RX == 0xfc) { //quality = buf; Serial.print("QV:"); Serial.print(buf, DEC); Serial.println(); Serial.println(); } else { } } } else if (startbyte == true) { //Waveform data } }


================================================================================

