Discussion Chain ID: 9359
================================================================================

User asked: Arduino unstable analog reading when using power supply

As soon as I unplug my Arduino from USB and only use the VIN pin with my 9 V power supply, the analog readings from my Arduino start to get unstable, which means the value fluctuates much more than with USB connected. When I reconnect USB again, the readings are stable again. I'm trying to build a load scale using a load cell, an INA125P and the Arduino, using this setup here: [Instrument Amplifier Calibration](https://learn.adafruit.com/smart-cocktail-shaker/calibration). Therefore I need stable readings. For example, I wrote some code to convert the `analogRead()` value to grams. While USB and VIN (9 V) connected, I get a stable reading of about 8.3 to 8.7 grams when I put a weight of 8.5 grams on the scale. As soon as I unplug USB, the LCD shows values from 7.8 grams up to 9 grams. Unfortunately, I don't have any other power supply which I could use to test. I've already tried to use a 9 V battery with the same results. Do I maybe need some additional de-coupling capacitors?

Discussion:
Comment: It's more likely that you need a beefier regulator. The on-board regulator on an Arduino can only supply a tiny amount of current (a few tens of mA) at 5V to external circuitry (beyond what the Arduino itself needs). When the USB cable is plugged in, the USB host can supply at least 500 mA (probably more) at 5V without any problem.
Comment: First of all what is your board? 'cause different boards have different regulators. And then... If you need precise values, avoid using the supply voltage as reference: it is really noisy and unprecise. Use the internal analog reference if it suits your needs, or otherwise use an external voltage reference. Use the power supply reference only when you don't need precision or you have a ratiometric sensor (e.g. a potentiometer)
Comment: I got the same issue where too much ripple from power supply, I saw this article bit late where I used an digital sensor to get away with analog read.

Answer 1 (5 votes):
I solved a similar issue by adding a pin from supply to AREF and it's now stable. This fixed a frustrating pH meter issue on my Mega that every time I plugged USB in it would be right, but as soon as I dropped the USB cable it changed the readings.

Discussion on this answer:
Comment: This answer was reported as "not an answer", I find that false. He just provided an example before his solution (rearranged that now). If his answer is not useful, please use the `This answer is not useful` button instead of the `Flag` button.
Comment: OK @Avamander. Understood.

Answer 2 (3 votes):
Load cell signals are very small and without proper shielding are be susceptible to noise. The USB power connector is likely earth grounded (the clue was battery operation is noisy, exact opposite of expectations!) I suggest testing this hypothesis: 1. Check with ohm meter if exposed metal on USB connector is connected to earth (chassis on your PC is usually earth connected). 2. Touch an earth grounded conductor to the signal ground of the load cell amplifier and see of the readings become more accurate. The final solution is to use properly shielded signal wires between the load cell and the amplifier. Also the signal wires between load cell and amplifier should be a 'twisted pair' to further reduce noise.

Answer 3 (2 votes):
I would say that this may be related to the analog reference. Try [AnalogReference](https://www.arduino.cc/en/Reference/AnalogReference) and see if it helps.

Answer 4 (0 votes):
Without much information (software, schematic, hardware, measurements ...) It is hard to tell. My guess would be that you used vcc as a dc reference and the circuit draws too much current from the battery than it can deliver so bcc dips, leading to a rise and instability in the reading. Good luck debugging it.


================================================================================

