Discussion Chain ID: 79628
================================================================================

User asked: Current Sensor ACS712 not changing its value even when connected bulb is lit

My task is to find the amount of current a light bulb uses, with that motive I used ACS712 current sensor of 30A connected all the connections properly as shown in the circuit diagram below [](https://i.sstatic.net/tFf0d.png) I tried all different codes available to calculate current... **Arduino Code :** void setup(){ Serial.begin(9600); } void loop(){ int rVal = 0; int sampleDuration = 100; // 100ms int sampleCount = 0; unsigned long rSquaredSum = 0; int rZero = 511; // For illustrative purposes only - should be measured to calibrate sensor. uint32_t startTime = millis(); // take samples for 100ms while((millis()-startTime) < sampleDuration) { rVal = analogRead(A0) - rZero; rSquaredSum += rVal * rVal; sampleCount++; } double voltRMS = 5.0 * sqrt(rSquaredSum / sampleCount) / 1024.0; // x 1000 to convert volts to millivolts // divide by the number of millivolts per amp to determine amps measured // the 20A module 100 mv/A (so in this case ampsRMS = 10 * voltRMS double ampsRMS = voltRMS * 10.0; Serial.println(ampsRMS); Serial.println(analogRead(A0)); } **Output** [](https://i.sstatic.net/APTKj.png) The output remains the same, even when the bulb is still lighting up. Thank You in advance :)

Discussion:
Comment: print the values of `rSquaredSum / sampleCount` and `voltRMS`
Comment: How much current does your bulb actually use? Or put differently: what percentage of 30A is that?
Comment: @timemage, on the picture it is 230 V / 100 W. it should make a difference in my experience. but it could be lost in calculation
Comment: That is in part what I'm thinking, but also to make sure his expectations are in line with what the sensor will actually do.
Comment: do 10 readings at less than 100 ms intervals between readings ... do not do any calculation with the values ... store them in an array ... after the readings are completed, print the values to serial
Comment: Try to tell us the different types of code and the tutorials you followed so we can better narrow down the answer.

Answer 1 (1 votes):
In a comment about `rZero = 511;`, you wrote: > For illustrative purposes only - should be measured to calibrate sensor. This may well be the cause of your problem. If, instead of 511, the “zero” of your measurement is around 445, then you end up measuring a non-existing DC current of about 3.23 A. When you turn on the light bulb, you add roughly 435 mA (100 W / 230 V) of AC current. The total RMS current you see is then: √(3.232 \+ 0.4352) ≈ 3.26 A Barely visible at the resolution at which you are printing the result. And this is assuming you are using a 100 W bulb, as shown in the diagram. These bulbs are getting hard to come by. If you are using anything less powerful, you may not see any change in the displayed value. As a quick and dirty fix, you may replace `511` by `445`. A better solution would be to compute the quadratic average of the AC part of the signal alone. In other words, compute: √(average(current2) − average(current)2)

Discussion on this answer:
Comment: how did you conclude that the zero is 445? the time between those measurements is more then 100 milliseconds. it could simply catch the same part of the wave
Comment: @Juraj: A constant reading yielding 3.23 A must be 511±66, but the values shown in the serial monitor are only consistent with 511−66.


================================================================================

