Discussion Chain ID: 78875
================================================================================

User asked: How can I measure maximum range of my ohm meter?

Hello everyone I am working on a project that will measure ohm value of a resistor . [Schematic / article link](https://create.arduino.cc/projecthub/iasonas-christoulakis/how-to-make-an-arduino-ohm-meter-90fda8%5DSchematic) <https://create.arduino.cc/projecthub/iasonas-christoulakis/how-to-make-an-arduino-ohm-meter-90fda8> And Here is the code below int analogPin= 0; int raw= 0; int Vin= 5; float Vout= 0; float R1= 1000; float R2= 0; float buffer= 0; void setup() { Serial.begin(9600); } void loop() { raw= analogRead(analogPin); if(raw) { buffer= raw * Vin; Vout= (buffer)/1024.0; buffer= (Vin/Vout) -1; R2= R1 * buffer; Serial.print("Vout: "); Serial.println(Vout); Serial.print("R2: "); Serial.println(R2); delay(1000); } } There need to be a known and another unknown resistor. The ohm value of known resistor is supposed to be 1k ohm So my question is, what is the max ohm range will my arduino ohm meter be able to measure of the unknown resistor with 1k reference resistor and how to calculate it? 5 - 10% accuracy is enough for me Thanks for giving me your attention :D

Discussion:
Comment: https://create.arduino.cc/projecthub/iasonas-christoulakis/how-to-make-an-arduino-ohm-meter-90fda8
Comment: Your question would make more sense if you gave an acceptable error percentage. You could read virtually any resistor, it just might be 100% wrong. Would 50% do? How about 10%? 1%?
Comment: @Nick Gammon almost 5% to 10%

Answer 1 (2 votes):
Your _resolution_ of the ADC (the analogRead) will be 1/1024 * 5V (since you get a reading in the range 0 to 1023), that is, you can tell the difference in increments of 5/1024 which is 0.00488V, so say roughly 0.005V. Now, I wrote a little bit of code to work out the voltages that you would get with various values for your varying resistor: r1 = 0, v = 5.000 r1 = 100, v = 4.545 r1 = 200, v = 4.167 r1 = 300, v = 3.846 r1 = 400, v = 3.571 r1 = 500, v = 3.333 r1 = 600, v = 3.125 r1 = 700, v = 2.941 r1 = 800, v = 2.778 r1 = 900, v = 2.632 r1 = 1000, v = 2.500 r1 = 1100, v = 2.381 r1 = 1200, v = 2.273 r1 = 1300, v = 2.174 r1 = 1400, v = 2.083 r1 = 1500, v = 2.000 r1 = 1600, v = 1.923 r1 = 1700, v = 1.852 r1 = 1800, v = 1.786 r1 = 1900, v = 1.724 r1 = 2000, v = 1.667 r1 = 3000, v = 1.250 r1 = 4000, v = 1.000 r1 = 5000, v = 0.833 r1 = 6000, v = 0.714 r1 = 7000, v = 0.625 r1 = 8000, v = 0.556 r1 = 9000, v = 0.500 r1 = 10000, v = 0.455 r1 = 20000, v = 0.238 r1 = 30000, v = 0.161 r1 = 40000, v = 0.122 r1 = 50000, v = 0.098 r1 = 60000, v = 0.082 r1 = 70000, v = 0.070 r1 = 80000, v = 0.062 r1 = 90000, v = 0.055 r1 = 100000, v = 0.050 r1 = 110000, v = 0.045 r1 = 120000, v = 0.041 r1 = 130000, v = 0.038 r1 = 140000, v = 0.035 r1 = 150000, v = 0.033 r1 = 160000, v = 0.031 r1 = 170000, v = 0.029 r1 = 180000, v = 0.028 r1 = 190000, v = 0.026 r1 = 200000, v = 0.025 r1 = 210000, v = 0.024 r1 = 220000, v = 0.023 r1 = 230000, v = 0.022 r1 = 240000, v = 0.021 r1 = 250000, v = 0.020 r1 = 260000, v = 0.019 r1 = 270000, v = 0.018 r1 = 280000, v = 0.018 r1 = 290000, v = 0.017 r1 = 300000, v = 0.017 You can see that the difference in voltage is only 0.005V (the ADC resolution) between r1 of 90k and 100k. Further down, you can't tell the difference (within 0.005V) of 270k and 300k. I'm not sure how to express that as an error percentage, perhaps someone better at maths than me can do that. :)

Discussion on this answer:
Comment: So does that literally mean that I can measure max 100k resistor with 1k reference resistor? Sorry to say but ur explanation has confused me a little bit.
Comment: I'm not saying that, but if you look at the voltage for 200k (0.025V) and 250k (0.020V), then you won't be able to tell the difference between 200k and (say) 220k. So you might say, for 200k that it would be +/- 50k. That is your error amount. So you could measure a 200k resistor, but the error amount would be 50/200, that is 25%. Now for 100k, it would appear to be 10/100, or 10%. Subject to my arithmetic being right, that would seem to indicate that 100k is about your limit, if you want 10% accuracy or better.
Comment: There would be a reason why cheap multimeters want you to choose a range for resistance measurement, they probably swap in an appropriate reference resistor to give a reasonable error amount for the range you are measuring.
Comment: Ok thanks , getting it clear now
Comment: Even the very expensive ohm meters switch range, it may be transparent to you but they do. I have repaired and calibrated many over the years and they all did it.
Comment: How? I suppose one way would be to have half a dozen reference resistors, and use a MOSFET to switch the signal being measured into one rather than the other (in other words, multiplex it).
Comment: @Nick, yes I am working on a auto ohm meter with my UNO , some MOSFET and resistors of 1k 22k 47k 100k and 220 k which's maximum range should be 1m+ . All u have to do is that, get values from reference resistor then logically edit it, to switch them automatically


================================================================================

