Discussion Chain ID: 77951
================================================================================

User asked: The analog read is not giving the correct waveform as that of an oscilloscope

I have code that fetches analog input values from terminals A0, A1 and A2 of an Arduino Mega 2560. When I plot the data, it does not give the desired waveform. The desired waveform is also measured with a the digital storage oscilloscope. I was thinking to make use of an Arduino Mega 2560 as an oscilloscope that can have 16 channels, because the DSO has only 4 and for a project, I urgently need 16 channels. Would be great if the community people suggest something. [](https://i.sstatic.net/llNe9.jpg) Below is the desired data from the oscilloscope that is measured in parallel to avoid any confusion about the circuit. [](https://i.sstatic.net/pe0Fm.jpg) %This is a script that will plot Arduino analogRead values in real time %Modified from http://billwaa.wordpress.com/2013/07/10/matlab-real-time-serial-data-logger/ %The code from that site takes data from Serial clear all close all clc %User Defined Properties a = arduino('Com5') % define the Arduino Communication port arduino=serial('COM5','BaudRate',9600); s1 = serial('COM5'); % Define here YOUR serial port s1.BaudRate=9600; plotTitle = 'Arduino Data Log'; % plot title xLabel = 'Time (s)'; % x-axis label yLabel = 'Voltage (V)'; % y-axis label legend1 = 'Voltage 1' legend2 = 'Voltage 2' legend3 = 'Voltage 3' yMax = 5 %y Maximum Value yMin = 0 %y minimum Value plotGrid = 'on'; % 'off' to turn off grid min = 0; % set y-min max = 5; % set y-max delay = .01; % make sure sample faster than resolution %Define Function Variables time = 0; data = 0; data1 = 0; data2 = 0; count = 0; %Set up Plot plotGraph = plot(time,data,'-r' ) % every AnalogRead needs to be on its own Plotgraph hold on %hold on makes sure all of the channels are plotted plotGraph1 = plot(time,data1,'-b') plotGraph2 = plot(time, data2,'-g' ) title(plotTitle,'FontSize',15); xlabel(xLabel,'FontSize',15); ylabel(yLabel,'FontSize',15); legend(legend1,legend2,legend3) axis([yMin yMax min max]); grid(plotGrid); tic while ishandle(plotGraph) %Loop when Plot is Active will run until plot is closed % dat = readVoltage(a,'A0')* 0.48875855327; %Data from the arduino % dat1 = readVoltage(a,'A1')* 0.48875855327; % dat2 = readVoltage(a,'A2')* 0.48875855327; dat = readVoltage(a,'A0')* 1; %Data from the arduino dat1 = readVoltage(a,'A1')* 1; dat2 = readVoltage(a,'A2')*1; count = count + 1; time(count) = toc; data(count) = dat(1); data1(count) = dat1(1) data2(count) = dat2(1) %This is the magic code %Using plot will slow down the sampling time.. At times to over 20 %seconds per sample! set(plotGraph,'XData',time,'YData',data); set(plotGraph1,'XData',time,'YData',data1); set(plotGraph2,'XData',time,'YData',data2); axis([0 time(count) min max]); %Update the graph %pause(delay); end delete(a); disp('Plot Closed and arduino object has been deleted');`

Discussion:
Comment: What sketch are you using on the Arduino? Are you using something like Firmata? Where does the `readVoltage()` function come from? It isn't anywhere in the link, where you modified the code from. From your images it is obvious, that you are sampling way too slow for seeing the wave form.
Comment: You are trying to sample at ≈ 12 samp/s a waveform that is around 3 kHz. You must read about the [Nyquist–Shannon sampling theorem](https://en.wikipedia.org/wiki/Nyquist%E2%80%93Shannon_sampling_theorem).
Comment: why are you sampling 3 channels? ... first determine if one channel can capture the data, then try two channels, etc.

Answer 1 (3 votes):
Your sampling rate is far too low for the signals you are looking at; you're not getting enough data points for drawing something that looks like the original signal. The Arduino library's `analogRead()`, while slow, can do better, up to about 9000 samples/s; the ADC in the atmega2560 in free-running mode can do about 75000 samples/s at 8 bits resolution. This would require programming it at the register level. There is only one ADC, so if you want 16 channels, this leaves about 4700 samples/s for each channel. In that configuration, however, the first measurements aren't valid and the second ones should be used, further reducing the sample rate per channel to about 2300 samples/s. With a perfect reconstruction filter, this would limit you to measuring signals up to only 1150 Hz. All this doesn't even consider the time it takes to store or transmit the results. So, an Arduino Mega 2560 will not do for making a usable 16-channel oscilloscope for your signals; it's not even close.


================================================================================

