Discussion Chain ID: 14364
================================================================================

User asked: Can I use micros() to analyze TX23 anemometer signal?

I'm very new to Arduino, and I'm trying to read data from a LaCrosse TX23 V02(X) anemometer. It's supposed to behave like this <https://www.john.geek.nz/2012/08/la-crosse-tx23u-anemometer-communication-protocol/>, but it's not... I don't have an oscilloscope, so I made this code, using micros() to try to trace the output from the TX23: void loop() { collectData(); } int collectData(void){ if (bitLength < 0){ bitLength = 1200;// length of bit is equal to approx 1230 micro seconds; } // init sequence pinMode(dataPin, OUTPUT); Serial.print("Send LOW\n"); digitalWrite(dataPin, LOW ); Serial.print("Send HIGH\n"); digitalWrite(dataPin, HIGH); delay(100); Serial.print("Send LOW\n"); digitalWrite(dataPin, LOW ); delay(500); Serial.print("Begin input mode\n"); pinMode(dataPin, INPUT); int state = 0; int oldstate = 0; unsigned long oldmicros = micros(); for (int i = 1; i<5250 ; i++) { state = (digitalRead(dataPin) == LOW)? 0:1; if (i%10 != 0) { Serial.print("-"); } else { Serial.print("+"); } delayMicroseconds(10); if (state != oldstate) { oldstate = state; i = 1; Serial.print(oldstate); Serial.print(" @"); Serial.print(micros() - oldmicros); oldmicros = micros(); Serial.print("\n"); } } Serial.print(state); Serial.println(micros()-oldmicros); } Does it have a chance to provide a quite accurate monitoring of the TX23 output? (eg is using micros() or delayMicroseconds(10) accurate ?)

Answer 1 (1 votes):
I'm going to present a third way of working on this. :) First, since I don't have a TX23 I wrote a TX23-simulator: const byte TESTPIN = 2; const byte test [6] = { 0b00000011, 0b01111011, 0b01010100, 0b00010100, 0b01001010, 0b10111111 }; void setup () { pinMode (TESTPIN, INPUT); delay (100); } // end of setup void loop () { // wait for host to pull pin low while (digitalRead (TESTPIN) == HIGH) { } // wait for host to release it while (digitalRead (TESTPIN) == LOW) { } digitalWrite (TESTPIN, HIGH); pinMode (TESTPIN, OUTPUT); for (byte i = 0; i < 6; i++) for (byte j = 0; j < 8; j++) { delayMicroseconds (1100); digitalWrite (TESTPIN, test [i] & bit (7 - j)); } pinMode (TESTPIN, INPUT); } // end of loop Based on your [linked page about the TX23](https://www.john.geek.nz/2012/08/la-crosse-tx23u-anemometer-communication-protocol/) it outputs a series of low/high pulses corresponding to the example on that page, like this: [](https://i.sstatic.net/pGCKr.png) Note that you need a 10 k pull-up resistor for this to work properly. * * * Now the reading code uses interrupts to detect state changes: const byte READ_PIN = 2; const byte NUMBER_OF_CHANGES = 48; volatile unsigned long bitTime [NUMBER_OF_CHANGES]; volatile byte state [NUMBER_OF_CHANGES]; volatile byte count; void stateChange () { unsigned long now = micros (); if (count >= NUMBER_OF_CHANGES) return; // array full state [count] = digitalRead (READ_PIN); bitTime [count++] = now; } // end of stateChange void setup () { Serial.begin (115200); Serial.println (); } // end of setup void takeReading () { count = 0; digitalWrite (READ_PIN, LOW); // start reading pinMode (READ_PIN, OUTPUT); delay (20); // pulse width pinMode (READ_PIN, INPUT); // release line EIFR = bit (INTF0); // clear flag for interrupt 0 attachInterrupt (0, stateChange, CHANGE); delay (55); // let readings fill up array detachInterrupt (0); } // end of takeReading void loop () { takeReading (); if (count == 0) return; // nothing! delay (500); for (byte i = 0; i < count; i++) { Serial.print ("count: "); Serial.print ((int) i); Serial.print (" = "); Serial.print (bitTime [i]); Serial.print (", state = "); Serial.print (state [i]); if (i > 0) { Serial.print (", width = "); unsigned long width = bitTime [i] - bitTime [i - 1]; Serial.print (width); Serial.print (" ("); int bits = width / 1000; for (byte k = 0; k < bits; k++) Serial.print (state [i - 1]); Serial.print (")"); } Serial.println (); } Serial.println (); } // end of loop * * * Results were: count: 0 = 10952804, state = 1 count: 1 = 10953928, state = 0, width = 1124 (1) count: 2 = 10960616, state = 1, width = 6688 (000000) count: 3 = 10962844, state = 0, width = 2228 (11) count: 4 = 10963960, state = 1, width = 1116 (0) count: 5 = 10968412, state = 0, width = 4452 (1111) count: 6 = 10969524, state = 1, width = 1112 (0) count: 7 = 10971752, state = 0, width = 2228 (11) count: 8 = 10972876, state = 1, width = 1124 (0) count: 9 = 10973988, state = 0, width = 1112 (1) count: 10 = 10975104, state = 1, width = 1116 (0) count: 11 = 10976216, state = 0, width = 1112 (1) count: 12 = 10977328, state = 1, width = 1112 (0) count: 13 = 10978440, state = 0, width = 1112 (1) count: 14 = 10984012, state = 1, width = 5572 (00000) count: 15 = 10985124, state = 0, width = 1112 (1) count: 16 = 10986244, state = 1, width = 1120 (0) count: 17 = 10987356, state = 0, width = 1112 (1) count: 18 = 10990700, state = 1, width = 3344 (000) count: 19 = 10991812, state = 0, width = 1112 (1) count: 20 = 10994040, state = 1, width = 2228 (00) count: 21 = 10995156, state = 0, width = 1116 (1) count: 22 = 10996268, state = 1, width = 1112 (0) count: 23 = 10997380, state = 0, width = 1112 (1) count: 24 = 10998500, state = 1, width = 1120 (0) count: 25 = 10999616, state = 0, width = 1116 (1) count: 26 = 11000728, state = 1, width = 1112 (0) This seems to agree with the test data.

Discussion on this answer:
Comment: Thanks for this, it really helps me progress and opens other possibilities to me.

Answer 2 (1 votes):
You should really make use of `pulseIn()` to work out the bit rate of the signal. The first HIGH pulse in the header is there for timing so you know how often to poll the input. Step one, after sending your outgoing initiation pulse, should be to wait for that first pulse (you might need a slight delay first before it starts transmitting - wait for the line to be low first): while (digitalRead(dataPin) == HIGH); // Wait for TX start uint32_t pulseWidth = pulseIn(dataPin, HIGH); // Get first pulse That gives you the width of the first pulse in microseconds. Now you can read 45 times at that period. If you first delay by half that amount before your first sample you will be roughly in the middle of the bit so you can cope with a slight amount of drift. // Wait for the middle of the first bit delayMicroseconds(pulseWidth / 2); uint32_t highBlock = 0; uint32_t lowBlock = 0; for (int i = 0; i < 45; i++) { // Shift the high 32-bits left one bit highBlock <<= 1; // Place in the highest bit of the low block as the lowest // bit of the high block highBlock |= (lowBlock & 0x80000000UL ? 1 : 0); // Shift the low 32-bits left one bit lowBlock <<= 1; // Add in the value of the bit from the data pin lowBlock |= digitalRead(dataPin); // Wait for the next bit delayMicroseconds(pulseWidth); } You should now have two 32-bit words which, when put together, form the entire datagram. You can now use bitwise operators to pull out the parts you are interested in. A better, though more complex, method of doing it would be to employ a timer with an overflow period equal to the pulse width. That will afford you much more accurate timing than a simple loop with delays.

Discussion on this answer:
Comment: Thanks a lot for the input. It does not exactly answer my initial question, but it certainly helps me solve my problem reading the TX23 output ! It works, but I don't understand the first 'while (digitalRead(dataPin) == HIGH);' and the data I get with pulseIn() are the first two bits of the header, not an initial pulse...
Comment: Marking this as right answer as it helped me resolve my real problem, but other two contributions were really helpful as well. Thanks to jwpat7 and Nick Grammon.

Answer 3 (1 votes):
Majenko showed a method for digitally recording the waveform, after measuring `pulseWidth` at the outset, and mentioned use of a timer for delays. I'll address the part of your question about using `micros()` while tracing TX23 output: “Does it have a chance to provide a quite accurate monitoring of the TX23 output?” Some of the serial output that your method produces as it goes along will affect evenness of timing. For example, the `Serial.print()` of + after every tenth reading introduces a small timing irregularity (the two branches of the `if` may differ a few cycles in execution time). More seriously, your double-read of `micros()` (after you print `oldstate`, you print `micros() - oldmicros` and then read `micros()` again) introduces a few cycles of timing error. It probably amounts to a few microseconds total over the course of recording the pulse train. Instead of reading `micros()` twice, say something like `newmicros = micros()` directly after detecting a state change, before doing any serial printing, then use `newmicros` in place of other `micros()` calls. Having serial prints before `oldmicros = micros();` causes an offset error, which will be microseconds or milliseconds depending on the data rate you set in `setup()`. It mostly affects the reported length only of the first pulse. What I would do for debug monitoring of the pulse train is along the following lines: enum { timelen=44, maxpause=50000 }; unsigned long int times[timelen]; int collectData() { byte ntimes = 0, state=LOW; unsigned long int prevtime; pinMode(dataPin, OUTPUT); digitalWrite(dataPin, LOW ); // Pull DTR low to trigger conversion delay(500); pinMode(dataPin, INPUT_PULLUP); // Release DTR and set pullup delay(1); // Let TX23U pull DTR low prevtime = micros(); while (micros() - prevtime < maxpause && ntimes < timelen) { if (digitalRead(dataPin) != state) { times[ntimes++] = prevtime = micros(); state = 1-state; } } return ntimes; } This code is intended to record times at which state changes occur. After `loop()` calls `collectData()`, it can call a `reportData()` routine that will print out pulse lengths by subtracting adjacent entries of `times[]`. Even-indexed array entries are times of rising edges in the pulse train; odd-indexed entries are times of falling edges. Because the `if (digitalRead(dataPin) != state) {...}` statement isn't executed atomically, the data pin can change after being tested but before being recorded a couple of lines later. Using `state = 1-state;` instead of `state = digitalRead(dataPin);` guarantees that even indexes are for rising edges, but does not guarantee that adjacent entries correspond to the same pulse edge. That is, a noisy signal can produce misleading results.


================================================================================

