Pregnancy Image Gallery
Pregnancy Image Gallery


Joe Raedle/Getty Images­

Introduction to How Ultrasound Works






Photo courtesy Karim and Nancy Nice­

What is Ultrasound?

In ultrasound, the following events happen:




An ultrasound machine


The Ultrasound Machine


A basic ultrasound machine has the following parts:

  • Transducer probe - probe that sends and receives the sound waves
  • Central processing unit (CPU) - computer that does all of the calculations and contains the electrical power supplies for itself and the transducer probe
  • Transducer pulse controls - changes the amplitude, frequency and duration of the pulses emitted from the transducer probe
  • Display - displays the image from the ultrasound data processed by the CPU
  • Keyboard/cursor - inputs data and takes measurements from the display
  • Disk storage device (hard, floppy, CD) - stores the acquired images
  • Printer - prints the image from the displayed data

­The transducer probe is th­e main part of the ultrasound machine. The transducer probe makes the sound waves and receives the echoes. It is, so to speak, the mouth and ears of the ultrasound machine. The transducer probe generates and receives sound waves using a principle called the piezoelectric (pressure electricity) effect, which was discovered by Pierre and Jacques Curie in 1880. In the probe, there are one or more quartz crystals called piezoelectric crystals. When an electric current is applied to these crystals, they change shape rapidly. The rapid shape changes, or vibrations, of the crystals produce sound waves that travel outward. Conversely, when sound or pressure waves hit the crystals, they emit electrical currents. Therefore, the same crystals can be used to send and receive sound waves. The probe also has a sound absorbing substance to eliminate back reflections from the probe itself, and an acoustic lens to help focus the emitted sound waves.

Transducer probes come in many shapes and sizes, as shown in the photo above. The shape of the probe determines its field of view, and the frequency of emitted sound waves determines how deep the sound waves penetrate and the resolution of the image. Transducer probes may contain one or more crystal elements; in multiple-element probes, each crystal has its own circuit. Multiple-element probes have the advantage that the ultrasounc beam can be "steered" by changing the timing in which each element gets pulsed; steering the beam is especially important for cardiac ultrasound (see Basic Principles of Ultrasound for details on transducers). In addition to probes that can be moved across the surface of the body, some probes are designed to be inserted through various­ openings of the body (vagina, rectum, esophagus) so that they can get closer to the organ being examined (uterus, prostate gland, stomach); getting closer to the organ can allow for more detailed views.

The parts of an ultrasound machine

The CPU is the brain of the ultrasound machine. The CPU is basically a computer that contains the microprocessor, memory, amplifiers and power supplies for the microprocessor and transducer probe. The CPU sends electrical currents to the transducer probe to emit sound waves, and also receives the electrical pulses from the probes that were created from the returning echoes. The CPU does all of the calculations involved in processing the data. Once the raw data are processed, the CPU forms the image on the monitor. The CPU can also store the processed data and/or image on disk.


3-D ultrasound images

Photo courtesy Philips Research

Different Types of Ultrasound

­The ultrasound that we ha­ve described so far presents a two-dimensional image, or "slice," of a three-dimensional object (fetus, organ). Two other types of ultras­ound are currently in use, 3-D ultrasound imaging and Doppler ultrasound.

In the past several years, ultrasound machines capable of three-dimensional imaging have been developed. In these machines, several two-dimensional images are acquired by moving the probes across the body surface or rotating inserted probes. The two-dimensional scans are then combined by specialized computer software to form 3-D images.

3-D imaging allows you to get a better look at the organ being examined and is best used for:

  • Early detection of cancerous and benign tumors (examining the prostate gland for early detection of tumors, looking for masses in the colon and rectum, detecting breast lesions for possible biopsies)
  • Visualizing a fetus to assess its development, especially for observing abnormal development of the face and limbs
  • Visualizing blood flow in various organs or a fetus

­Doppler ultrasound is based upon the Doppler Effect. When the object reflecting the ultrasound waves is moving, it changes the frequency of the echoes, creating a higher frequency if it is moving toward the probe and a lower frequency if it is moving away from the probe. How much the frequency is changed depends upon how fast the object is moving. Doppler ultrasound measures the change in frequency of the echoes to calculate how fast an object is moving. Doppler ultrasound has been used mostly to measure the rate of blood flow through the heart and major arteries.


Major Uses of Ultrasound

Ultrasou­nd has been used in a variety of clinical settings, including obstetrics and gynecology, cardiology and cancer detection. The main advantage of ultrasound is that certain structures can be observed without using radiation. Ultrasound can also be done much faster than X-rays or other radiographic techniques. Here is a short list of some uses for ultrasound:

Obstetrics and Gynecology
  • measuring the size of the fetus to determine the due date
  • determining the position of the fetus to see if it is in the normal head down position or breech
  • checking the position of the placenta to see if it is improperly developing over the opening to the uterus (cervix)
  • seeing the number of fetuses in the uterus
  • checking the sex of the baby (if the genital area can be clearly seen)
  • checking the fetus's growth rate by making many measurements over time
  • detecting ectopic pregnancy, the life-threatening situation in which the baby is implanted in the mother's Fallopian tubes instead of in the uterus
  • determining whether there is an appropriate amount of amniotic fluid cushioning the baby
  • monitoring the baby during specialized procedures - ultrasound has been helpful in seeing and avoiding the baby during amniocentesis (sampling of the amniotic fluid with a needle for genetic testing). Years ago, doctors use to perform this procedure blindly; however, with accompanying use of ultrasound, the risks of this procedure have dropped dramatically.
  • seeing tumors of the ovary and breast
  • seeing the inside of the heart to identify abnormal structures or functions
  • measuring blood flow through the heart and major blood vessels
  • measuring blood flow through the kidney
  • seeing kidney stones
  • detecting prostate cancer early

In addition to these areas, there is a growing use for ultrasound as a rapid imaging tool for diagnosis in emergency rooms.


There have been many concerns about the safety of ultrasound. Because ultrasound is energy, the question becomes "What is this energy doing to my tissues or my baby?" There have been some reports of low birthweight babies being born to mothers who had frequent ultrasound examinations during pregnancy. The two major possibilities with ultrasound are as follows:

  • development of heat -- tissues or water absorb the ultrasound energy which increases their temperature locally
  • formation of bubbles (cavitation) -- when dissolved gases come out of solution due to local heat caused by ultrasound

However, there have been no substantiated ill-effects of ultrasound documented in studies in either humans or animals. This being said, ultrasound should still be used only when necessary (i.e. better to be cautious).

An Ultrasound Examination

For an ultrasound exam, you go into a room with a technician and the ultrasound machine. The following happens:

  1. You remove your clothes (all of your clothes or only those over the area of interest).
  2. The ultrasonographer drapes a cloth over any exposed areas that are not needed for the exam.
  3. The ultrasonographer applies a mineral oil-based jelly to your skin -- this jelly eliminates air between the probe and your skin to help pass the sound waves into your body.
  4. The ultrasonographer covers the probe with a plastic cover.
  5. He/she passes the probe over your skin to obtain the required images. Depending upon the type of exam, the probe may be inserted into you.
  6. You may be asked to change positions to get better looks at the area of interest.
  7. After the images have been acquired and measurements taken, the data is stored on disk. You may get a hard copy of the images.
  8. You are given a towelette to clean up.
  9. You get dressed.

The Future of Ultrasound

­As with­ other computer technology, ultrasound machines will most likely get faster and have more memory for storing data. Transducer probes may get smaller, and more insertable probes will be developed to get better images of internal organs. Most likely, 3-D ultrasound will be more highly developed and become more popular. The entire ultrasound machine will probably get smaller, perhaps even hand-held for use in the field (e.g. paramedics, battlefield triage). One exciting new area of research is the development of ultrasound imaging combined with heads-up/virtual reality-type displays that will allow a doctor to "see" inside you as he/she is performing a minimally invasive or non-invasive procedure such as amniocentesis or biopsy.

For more information on ultrasound, see the Links section.